Maximum and minimum setpoint values that can be set with outlet adjustment.B.: For low temperature units, the minimum value that can be set for the low limit is equal to the antifreeze setpoint 2 C (also check the value set for forced shut-down.
In the W3000, the message Com.
The fans may be turned on using the on/off system ( step ) in which the fans are turned on progressively one after the other (figure.3a) or mutually excluded (figure.3b).
Starts per hour 10 Min.If Off continues to be displayed it means that the unit has been switched.The Off from time bands message appears in the main mask to show that the unit has been switched off.Press UP and down to set the required keypad address (e.g.: 21).The self-adapting algorithm is able to measure system dynamics and calculate the minimum dead area in order to respect compressor activation times and the maximum number of start-ups per hour.If the plan network does not comprise just foxit reader 2.0 portugues W3000 chillers, a 16-key keyboard must be used.Access mask to freecooling submenu.Control Chiller Setpoint.0 bar Diff bar Cont.Slave N Value bar.Mask for checking connection with the slave expansion boards.Maximum and minimum variation limits can also be set.Expansion "E" - address 1 Digital inputs Digital inputs ID1 Recovery flow switch ID1 ID2 Recovery remote enable ID2 ID3 ID3 ID4 ID4 ID5 ID6 Analogue inputs ID7 Circuit 1 liquid temperature B1 ID8 Circuit 2 liquid temperature B2 ID9 Recovery inlet temperature B3 ID10.
Display address Setting.:00 I/O Board address: Display address Setting.:21 I/O Board address:- 3) This mask means that the address of the keyboard has been set.
Exp2 offset Value B7:.0.4 C B8: - - Master calibr.
Slave exp4 N Value 3 A.
Exp2 offset Value B3:.0.6 C B4:.0.5 C Modifies the probe offsets of expansion.Access mask to manufacturer menu.Sets the cycle time of this function, that is, the maximum time in the chiller plus recovery mode.If the pressure falls below this value, defrosting is forced Evaporation pressure values at which the circuit with more than one hermetic compressor per circuit is "separated" to return the operating point within limits Evaporation pressure value at which compressor "separation" terminates.If the unit is switched Off, it will remain in the Off from time bands mode.Either the board must be addressed or the plan network must be configured.4 adj.:000 does not apply).Master N Value C bar.Displays condenser inlet CDO1 CDO2 EHt OPT HP1 temperature (In water-water units, if the condenser outlet probe is enabled).
Default values Deletes the memory and installs "logical" values.
Access the setpoint menu and display the Operating mode parameter.
Only appears for units with two pumps.